The PC as a Medical Device
It is very convenient to design medical devices based on the PC hardware platform. However care must be taken to ensure that the resulting
product complies with the requirements of IEC 60601-1.
A standard PC power supply almost certainly does not comply with the electrical
IEC 60601-1 requirements from several standpoints, e.g.
leakage current requirements, dielectric strength requirements. There are two possible solutions:
, which has been designed to IEC 60601-1.
- Replace the standard PC power supply with a power supply which has been designed to comply with IEC 60601-1.
- Power the PC via a 1:1
The second route is preferable if there are additional pieces of the equipment (e.g. computer screen), which also need powering, as they also can receive their power via
the isolation transformer. The first route is advantageous from the standpoint of size and weight.
Note that in any case the PC used should be approved to the appropriate
safety standard for IT equipment, i.e. IEC 60950, or its national variants, EN 60950, UL 1950…
Simply powering a PC via a medical grade isolation transformer or an IEC
60601-1 approved power supply is not sufficient to ensure that the device complies with all the requirements of IEC 60601-1.
Some additional points are noted below:
If there is a requirement for the product to prevent the ingress of liquids, then the PC used would have to comply with the appropriate level of prevention of ingress of
- Patient connections, particularly Type BF and Type CF, must be isolated from other secondary circuitry used for signal input/output ports. This protects the patient if, as a
result of faults on external equipment, dangerous voltages appear on communication lines.
- Any communication ports which are not part of the medical device, should be blocked off so that they cannot be used.
- Connectors on patient leads must not be of a type in common use, e.g. 9-pin D-type, keyboard connector, etc., to prevent the possibly of the patient being inadvertently
connected to an unsafe device or port.
- The PC must not have red or yellow indicator lights, buttons or switches, as these have special meaning
In order to ensure that the device is fully compliant with all the requirements of IEC 60601-1 or UL 60601-1, it must be submitted to an accredited test laboratory